Mote 12 was the one that stopped working when initially installed at rsw02, but it got installed on the radiation stand at rsw04 to cap the soil sensor cables. I installed mote 8 in its place to bring it back to EOL for testing.
No progress on the rsw04 tower. The tower sections are still laying on the ground.
Tested mote12 back in Boulder. It works fine. However, found that all the screw terminal connections to the console port (where we inject power) were loose – some by many turns on the screw. This easily could have caused the power-up-but-then-stop problem we observed. Mote12 is now back in our pile of stuff to return to Portugal.
This suggests that checking these screw terminals should be part of our "prepping" for each station.