Sonics and gas analyzers
The plan is to replace it with the last RMYoung. 2% bad samples all the time would be better than 2% good samples during the day.
CSAT3A on port 2 is reporting all bad samples as of 2017-06-02,08 UTC. So we only have to replace whichever of head or box is bad, or both, but we have spares. Cycling power did not help.
Lots of flagged samples still. This is a CSAT3 IRGA. Up to %30 of the winds are flagged in qctables, need to check on the gas diagnostic.
Diagnostic still mostly bad.
Still needs to be replaced.
Isabel and I installed a second solar panel at tse05 around 16z. The panel is more horizontal and is aimed more west, since the first panel is aimed more east. It Looks like voltage jumped up a tenth, but maybe there will not be enough charging left today to get it through tonight. Maybe tomorrow night.