Weather: windy and rainy this morning. May clear up in the afternoon.
Sensor Status Overview
A number of sonics, ~7, were down this morning. It appears rain was the reason. Most are back up.
Three Licors were down but I was able to get running again via PIO commands.
TRH 10m and 60m on rsw06 were down due to fan current going to high. I was able to get them back
up via PIO. The max current cutoff has been increased, will monitor.